Mobilett Mira Max RDSR is classified as RF, while it is a DX system.
This is due to the use of TID 10004 AND TID 10007 by this system.
It seems there is no way to see that this system is a DX system. The device model name even reports as "Fluorospot Compact FD", although it isn't. I think the only option is a method like described in my comment in issue #445 (let the user have the option to set the modality type of the system).
I had forgotten about this conversation we had last year, sorry!
I have since changed the RDSR import routine to cater for the RDSRs produced by Canon and Carestream (ref #464), but looking at the example Siemens Mobilett RDSR I have I can see it wouldn't work with the new code.
I shall have to add it to the tests, but I think it will only be useful when ref #445 is complete.