Siemens mammography rdsr objects (available in Siemens Revelation systems) contain some errors that prevent correct viewing.
- Laterality code meaning is incorrect
code meaning is “Left Breast” / “Right Breast” and should be “Left breast” / “right breast” (shouldn’t be a capital B)
- For tomo-acquisitions the dose of one projection is given instead of the dose for all projections
So, actually I think both issues should be fixed by Siemens.
In the meanwhile we could change mgfiltered.html and mgdetail.html to compare case-insensitive (and actually I think it would be better to compare to code value instead of code meaning, although it would be less readable).