The Anatomical structure sequence doesn't appear to exist in the Siemens or Toshiba fluoro RDSRs
Fixed 3 TID 10003 elements that have not previously been used as the extraction code didn't match what would be found in the wild or match the models! Mammo RDSR now imports but thinks it is a fluoro study... Refs #368
Mammo RDSR should only complete the accumulated mammo xray dose table, not the others. Refs #368. acuisition_device_type should have been a ContextID foreignkey not a charfield. The value is currently never used, but should be to determine which Accumulated template should be expected. Created new cid version.
Filtering order, study delete rf, used the wrong DAP and Dose RP totals; corrected. Removed DAP unit conversion function from AccumProjXRayDose model - it shouldn't be used. Changed a couple of references to it. Refs #368
Laterality at event level was overwritten by anatomical structure value due to error in code. Fixed. CT and Fluoro don't generally use this as far as I know, so shouldn't be an issue. Updated detail view to include breast laterality in view column. Refs #368
Tested this branch against current develop: imported various fluoroscopy RDSRs into a fresh database using both branches, then exported them using csv and XSLX. Results were identical, though the version on this branch was faster on import and export. XLSX export in particular could do with a lot of optimisation though!